No, holding the button down doesn't work. I should add that even
though there are no "inlet pumps" in this system, there is a button
for them. It has always lit up when activated, but not now. So, it is
both the "source pumps" and "inlet pumps" that don't activate.
I tried venting, but this didn't change anything.
At 01:04 PM 14/08/2007, you wrote:
>Will it come on if you press and hold the on button? Those switches are
>notorious. Sometimes it a took a few tries. Each time but we had to hold it in
>until it "caught." The solution is solder a new switch in or hope
>each time that
> it would catch and hold.
>Quoting Paul Eby <[log in to unmask]>:
> > Hello all,
> > I'm looking for assistance with a problem after a power outage: the
> > pumps on our Delta Plus XL will not turn back on.
> > When I toggle the source pump button, nothing happens (LED light does
> > not come on, no pump activity). The instrument does indeed have power
> > (various fans and electronics are humming), and I have reset the
> > breakers on the back.
> > The roughing pump itself is fine - it works fine when plugged into a
> > hot socket. I've also gone through the system looking for blown fuses
> > and loose cables. I've also left it unplugged for a few hours hoping
> > for a miraculous reset of some sort. No luck.
> > Any ideas?
> > Paul Eby
> > [log in to unmask]
> > Biogeochemistry Lab (E-Hut, room 102)
> > School of Earth and Ocean Science
> > University of Victoria
> > PO Box 3055
> > Victoria, B.C.
> > Canada
> > V8W 3P6
> > phone: 250-721-6183
> > fax: 250-472-4620
>Greg Eischeid--Harvard University
>20 Oxford Street
>Cambridge, MA 02138 USA