I don't know where you can get a used S/L sampler but one from a Carlo Erba
NA1500 will fit, and is probably easier to come by. Surprisingly the
thread on the top of both the Europa SL and Carlo Erba 1500 is the
same. However, the lines going into the Carlo Erba are 2 mm OD, the lines
from the Europa are 1/8 inch. However, we discovered it was easy to make
an adapter by inserting a piece of 2 mm tube into a 1/8 inch tube and
soldering them together.
However, I have found the that our Europa sampler does have a smaller blank
for N than our 1500 NA sampler, so the Europa sampler is preferable.
At 04:44 PM 11/8/01 -0500, you wrote:
>Does anyone have a spare autosampler for a S/L Prep System for the
>Europa 20/ 20 Mass Spectrometer?
>If by chance one exists out there somewhere, and you would be willing to
>part with it (even temporarily) could you please contact Danielle
>Schmitt at [log in to unmask] or at 609-258-1432. If you personally
>do not possess this part but have any suggestions as to whom we should
>contact regarding such a search, we would be most appreciative!
>Thank you for your time,
Paul D. Brooks,
147 Higard Hall MC 3110,
UC Berkeley, Ca. 94720-3110.
[log in to unmask]