Do you mean that the start and stop positions of the software switch need
to be swapped? If so, I had this problem with our pre-Conflo I EA
interface. It can be changed through the software. E-mail me directly if
you want to try this, and I'll see if I can't remember how I did it.
[log in to unmask]
Biogeochemistry Lab (E-Hut, room 102)
School of Earth and Ocean Science
University of Victoria
PO Box 3055
At 06:51 PM 7/7/2003 -0700, you wrote:
>>Help ISOGEOCHEM gang...
>>I am running a Finnigan delta plus XL coupled to a CE EA1108 that is
>>outfitted with a Costech zero blank autosampler. I am having some
>>difficulties with my conflo II device, that have arisen after installing
>>ISODAT NT. Although we have identified what is causing the problem, I was
>>hoping someone out there had been through this as well, and would be able to
>>suggest a quick solution. Essentially, the contact closure that gives our
>>EA a signal to begin a cycle remains "closed" at rest, rather than only
>>closing for the second programmed by ISODAT. I have (blissfully) little
>>experience mucking around in the conflo II guts, but there does appear to be
>>a parity switch sitting in the interface. Has anyone found a need to
>>physically switch these leads after installing ISODAT NT? Or has anyone had
>>similar problems, but arrived at another solution?
>>Thank you in advance,
> NASA/Ames Research Center
> Mail Stop 239-4
> Moffett Field, CA 94035
> 650-604-1430 (voice mail)
> 650-604-6110 (lab)
> 650-604-1088 (fax)
>Incoming mail is certified Virus Free.
>Checked by AVG anti-virus system (http://www.grisoft.com).
>Version: 6.0.495 / Virus Database: 294 - Release Date: 6/30/2003
Outgoing mail is certified Virus Free.
Checked by AVG anti-virus system (http://www.grisoft.com).
Version: 6.0.495 / Virus Database: 294 - Release Date: 6/30/2003