Dear Eurovector LAS users,
have you already have something similar ?
since one week, I started to analyse 18O in urine samples with EA
EuroPyrOH/IRMS Isoprime. But the LAS gives problems :
the plunger of the SGE plunger-in-needle syringe is "busy"
after around 50 injections of urine sample (which are treated by activated
The resistence of the plunger in the needle stops the LAS during sequence.
So does it mean we must change the syringe type ?
or the sample treatment ?
Moreover since yesterday, the LAS stopped itself during sequences
(after one or two injections).
The screen switched off and restarted with Tray identification.
I have tested the fuses and the connexions, which are good.
Any help would be greatly appreciated.
CHU Hotel Dieu
3 étage aile Nord
Place Alexis Ricordeau
44093 Nantes cedex 1
Tel : 00 33 02 40 08 75 40
Fax : 00 33 02 40 08 75 44