I know that there has been communication on this forum about a similar
problem before, but all the solutions that have been giving do not seem
to help with my setup (or I am missing something).
I have installed Isodat 2.0 and Service pack 2.63 on a new computer that
it connected to the TC/EA and DeltaPlusXP. The Pal Loader is also
installed and with this software I could see that the GC Pal is
connected to COM1 and that they are communicating. In the configuration
I connected the Conflo-liquid (also COM1) to the configuration. In the
Time Event List there is a column for start autosampler and one for
Elemental analyser on. I gave a start autosampler command at 60 seconds
in the list. During a run I can see that Isodat gives a command (Inject
sample 1), but the GC Pal remains silent.
I hope somebody can tell me what I should do (or try). I read stuff on
Isogeochem about changing computer langues in scripts or changing from
acquisition to acquisition-old in the method, but I am not sure if this
also applies for the newest service pack. Please let me know if you have
had the same experience.
Free University Amsterdam