This is in regards to a change in carrier flow rate from our Flash EA 1112
to a Conflo III.
The carrier flow is set at 100ml/min. However, when the He dilution
capillary is dropped in, the flow rate drops to c. 58ml/min. I have checked
the heights of the capillaries in the open split and they are positioned
correctly according to the manual. Can anyone tell me why this happens?
Southern Cross University
Lismore NSW Australia