This used to happen on my old 252, but with a physical chart recorder
connected directly to the mass spec I could see that the signal was still
present coming out of the amp. Rebooting the computer reset the data feed.
[log in to unmask]
Alberta Innovates - Technology Futures*
Vancouver Island Technology Park
3 - 4476 Markham St.
Victoria, BC, Canada V8Z 7X8
*The Alberta Research Council, Alberta Ingenuity Fund, iCORE & nanoAlberta
are now part of Alberta Innovates - Technology Futures.
From: Stable Isotope Geochemistry [mailto:[log in to unmask]] On Behalf
Sent: June 27, 2011 8:34 AM
To: [log in to unmask]
Subject: [ISOGEOCHEM] signal problem
I've run into a problem I have to admit, I've never encountered. We lost
mass 44 voltage last night on our 253, this in the middle of a Kiel run, but
have retained M45 + M46 signal intensity. Can anyone point me in the right
direction as to what the problem might be?
KPESIL-Keck Paleoenvironmental & Environmental Stable Isotope Laboratory
University of Kansas, Dept. of Geology Multidisciplinary Research Building
2030 Becker Drive, RM 130 Lawrence, KS