Dear list members,
does anybody know if it's possible to change the Peak Center Pre-Delay time
in Isodat 2.0?
Whenever I run a sequence I cannot mark the "Peak Center" column, since the
software always returns an error message. The problem is that the Pre-Delay
time is set at 5 sec and within that time the signal from the Ref Gas does
not reach the MS, therefore no peak center takes place and the sample is not
injected. I know that I can do the peak center manually before the sequence
and then not mark the "Peak Center" column, but it would be nice to be able
to do it automatically before each run again.
I looked at the ISL scripts but could not find the one that I could edit.
Thanks in advance,
MARUM Zentrum f. Marine Umweltwissenschaften
Leobener Str. MARUM-Building, office 2570
[log in to unmask]