Herb and Isodat users,
There is another solution to this same kind of problem as well. Sometimes the peak center settings are too fast and the software fails to register the peak, even though it was found. To solve this problem, go to gas configuration editor, and select the peak center option by right clicking on the gas in question. Then, change the HV speed by changing the steps to move the HV more slowly. Hope this helps.