_________Remove the belt from the potentiometer, wind the pot back several turns and try again.
From: Stable Isotope Geochemistry [[log in to unmask]] on behalf of Pack, Andreas [[log in to unmask]]
Sent: Friday, January 08, 2016 5:48 AM
To: [log in to unmask]
Subject: [ISOGEOCHEM] MAT253 bellows hardware calibration
apart of having the annoying problem with the slipping rubber belt (that we are dealing with for the past years); since before christmas the “Hardware calibration” does not work anymore. When doing hardware calibration (iIrrespective of doing it in the Acquisition or Instrument control window); bellows are not moving and software freezes. We are using Isodat 3.
The signal calibration still works well after adjusting the bellows potentiometer position manually.
Did any of you had a similar problem (and a solution)?
Prof. Dr. Andreas Pack
Tel.: (Ms. Andrea Beckmann, secretary) +49 551 397901
Fax: +49 551 3910452
Tel: (office) +49 551 3912254, (mobile) +49 175 2981638