Have you run a scan for the extraction plates (intensity vs. extraction and sym)? Unless you get a nice curve it could be an op amp on the source control board went out. It would look like a nice curve at first, but flatten out too soon. Check all the plates to be sure. This has happened to me more than once.
Best of luck,