We want to buy Elemental Analyzer for our EA-IRMS analyses.
We know ISOPRIME, SERCON, THERMOSCIENTIFIC and EUROVECTOR.
Do you have any recommendation and alternative brand name?
Chemical Engineer, M.Sc.
Turkish Petroleum Corporation
Research Center – Geochemistry Department
Mustafa Kemal Mah. 2180. Cad. No.86
06100 Çankaya / Ankara / TURKEY
Phone : + 90 (312) 207 25 19
Fax : + 90 (312) 286 78 08
Are you looking for inorganic (lab) standards to preserve IAEA-N1 and IAEA-N2 (both ammonium sulphates) or are you looking for organic (lab) standards?
If it is the former you after you could purchase USGS25 and USGS26 (both ammonium sulphates too) with d15N values of -30.41 and +53.75 per mil, respectively.
If you require (lab) standards for scale calibration and quality control of organic samples I would recommend getting started by purchasing USGS40 and USGS41 (both glutamate) with d15N values of -4.52 and +45.57 per mil, respectively. Both these materials are recognised reference materials so you could use these to determine scale calibrated d15N values of lab standards, for which amino acids (as suggested by Stan) would be a good choice since their chemical make-up (in particular their C/N ratio) is a good match for most organic materials.
As also mentioned by Stan, some AAs are a pain to weigh out (either lumpy or electrostatically charged like the dickens). In these instances, we prepare aqueous stock solution which we dilute to end up with solutions where for example 10 microL contain 0.4 mg. In line with the desired sample weight, we dispense small volumes of the dilution into tin capsules sitting in a well-plate made of solid metal. This metal well-plate is eventually placed in a tray filled with liquid N2 to flash-freeze the samples and the frozen well-plate is subsequently placed in a freeze dryer. This procedure yields standards of highly reproducible sample weight (and, hence, N2 and CO2 peaks).
in order to preserve the IAEA standards N1 and N2 we like to have labstandards with delta 15 values of 0 ‰ and around 20 ‰. The zero value doesn't appear to be a problem, but a material with the higher value we can't find until now.
Does anyone of you find a readily available material with a d15N of approx 20‰?
Centre for Isotope Research, University of Groningen.
Centrum voor IsotopenOnderzoek (CIO), Rijksuniversiteit Groningen
Nijenborgh 4, 9747 AG Groningen, Netherlands
tel +31-50-3634716 fax +31-50-3634738
This email is from the James Hutton Institute, however the views
expressed by the sender are not necessarily the views of the James Hutton
Institute and its subsidiaries. This email and any attachments are confidential and
are intended solely for the use of the recipient(s) to whom they are addressed.
If you are not the intended recipient, you should not read, copy, disclose or rely on
any information contained in this email, and we would ask you to contact the
sender immediately and delete the email from your system. Although the James
Hutton Institute has taken reasonable precautions to ensure no viruses are present
in this email, neither the Institute nor the sender accepts any responsibility for any
viruses, and it is your responsibility to scan the email and any attachments.
The James Hutton Institute is a Scottish charitable company limited by guarantee.
Registered in Scotland No. SC374831
Registered Office: The James Hutton Institute, Invergowrie Dundee DD2 5DA.
Charity No. SC041796
Kisiye özel bu mesaj ve içerigindeki bilgiler gizlidir. Mesaj içeriginde bulunan bilgi, fikir ve yorumlar, sadece göndericiye aittir. TPAO bu mesajin içerigi ve ekleri ile ilgili olarak hukuksal hiçbir sorumluluk kabul etmez. Yetkili alicilardan
biri degilseniz, bu mesajin herhangi bir sekilde ifsa edilmesi, kullanilmasi, kopyalanmasi, yayilmasi veya mesajda yeralan hususlarla ilgili olarak herhangi bir islem yapilmasinin kesinlikle yasak oldugunu bildiririz.Böyle bir durumda lütfen hemen mesajin
göndericisini bilgilendiriniz ve mesaji sisteminizden siliniz.Internet ortaminda gönderilen e-posta mesajlarindaki hata ve/veya eksikliklerden veya virüslerden dolayi mesajin göndericisi herhangi bir sorumluluk kabul etmemektedir.
*** Bu mail zararli icerige karsi, TPAO Antivirus Sistemleri tarafindan taranmistir. ***
The information contained in this communication may contain confidential or legally privileged information. Responsibility about sent contents is belong to sender.TPAO doesn't accept any legal responsibility for the contents and attachments
of this message. If you are not the intended recipient you are hereby notified that any disclosure, use, copying, distribution or taking any action in reliance on the contents of this information is strictly prohibited. If you have received this communication
in error, please notify the sender immediately by responding to this e-mail and then delete it from your system. The sender does not accept any liability for any errors or omissions or any viruses in the context of this message which arise as a result of internet
*** This mail was scanned for known viruses by TPAO Antivirus systems.***