Hi Greg, It is a simple fix. In the method you can do 5 Fill Strokes with a 60 second delay and you’ll have the 5 minutes. I just checked our method and it has 10 fill strokes with a 60 second delay. Best, Brad Yale Analytical and Stable Isotope Center Yale Institute for Biospheric Studies 21 Sachem Street New Haven, CT 06511 [log in to unmask] 203-432-3189 > On Jan 11, 2017, at 4:13 PM, Michalski, Greg M <[log in to unmask]> wrote: > > We are trying to do a basic headspace analysis using a thermo gas bench and PAL. We cant find the parameter in the Method to keep the needle in the vial. We use the needle up delay in Method, but that has a max time of 99 S and we want 5 min or so to do multiple injections.