It is a simple fix. In the method you can do 5 Fill Strokes with a 60 second delay and you’ll have the 5 minutes. I just checked our method and it has 10 fill strokes with a 60 second delay.
Yale Analytical and Stable Isotope Center
Yale Institute for Biospheric Studies
21 Sachem Street
New Haven, CT 06511
We are trying to do a basic headspace analysis using a thermo gas bench and PAL. We cant find the parameter in the Method to keep the needle in the vial. We use the needle up delay in Method, but that has a max time of 99 S and we want 5 min or so to do multiple injections.