And the winner is . . .
Installed first extraction lens 90 degrees off.
Am I embarrassed? Yes. Will I ever do that again? No.
Though my mistake was amateurish, I learned a number of very useful things from your responses. Thanks everyone for your help.
I recently cleaned the source in our Delta V Plus. I also replaced the slit which was quite eroded. On reinstalling the clean source, the sensitivity for both C and N is about a quarter of what it used to be. I’ve tried three different filaments,
including the one that was running happily before the cleaning, all with the same results. I’ve tried manual tuning but I don’t get results any better than the auto-tune does. I’ve disassembled and reassembled again checking each ceramic piece and lead. The
extraction plates are oriented correctly. The stability of the diminished signal is very good (standard deviation ~ 0.02). We’ve cleaned the source previously with no problems.
Does this problem sound familiar to anyone? What might I be overlooking? All suggestions are welcome.
Michael Tuite, PhD
Jet Propulsion Laboratory
California Institute of Technology
[log in to unmask]
818 354-1247 office
626 353-1958 cell