Hi Greg, Running solids for 2H abundance analysis is or should be as straight forward as is running samples for 18O. Paired reference materials needed and available for scale normalisation of measured d2H values to the VSMOW/SLAP scale include solid compounds that were characterised by Arndt Schimmelmann’s international project, and reference waters in cold-welded silver tubes developed by Haiping Qi and Ty Coplen. All this materials are available from the Reston lab of the USGS. https://pubs.acs.org/doi/10.1021/acs.analchem.5b04392 https://www.tandfonline.com/doi/full/10.1080/10256016.2018.1538137 https://isotopes.usgs.gov/lab/referencematerials.html Where it gets “tricky” is the issue of analysing halocarbons or N-rich organics for 2H abundance because side-reactions during thermolysis mean H2 yield is not quantitative which in turn yields artefact d2H values due to associated mass discriminatory effects. For “recalcitrant “ compounds of that nature the TC/EA reactor has to be packed with either a mixture of glassy carbon chips and Cr granules or Cr granules only. There are number of papers by Matthias Gehre and colleagues describing this set-up in detail. Information can also be found in this phantastic book (coughs): https://onlinelibrary.wiley.com/doi/book/10.1002/9781119080190 Hope this helps. Best, Wolfram ****************************************************** Prof. Dr W Meier-Augenstein, CChem, FRSC Stable Isotope Forensics & Analytical Sciences Robert Gordon University School of Pharmacy and Life Sciences The Ian Wood Building Garthdee Road Aberdeen AB10 7GJ E-mail: [log in to unmask] ________________________________ From: Stable Isotope Geochemistry <[log in to unmask]> on behalf of Cane, Gregory A <[log in to unmask]> Sent: 30 September 2019 15:42:21 To: [log in to unmask] Subject: [ISOGEOCHEM] Ice core water and HD of solids Would anyone be willing to send us either Antarctic or Arctic ice core water they don't need, say 2 liters? We'll pay for shipping obviously. I recall some labs sometimes give this stuff away. While I'm on the subject, we're trying to set up a method to run HD of solids on our TCEA. I've run solids for d18O via TCEA, but not HD, and I recall something about HD solids analysis not being so straightforward, if that's accurate. So I'm looking for what works out there. Cheers, Greg ________________________________ Robert Gordon University has been awarded a TEF Gold award for the quality of its undergraduate teaching and learning, placing it in the top 20% of Universities in the UK Robert Gordon University, a Scottish charity registered under charity number SC 013781. This e-mail and any attachment is for authorised use by the intended recipient(s) only. It may contain proprietary material, confidential information and/or be subject to legal privilege. It should not be copied, disclosed to, retained or used by, any other party. If you are not an intended recipient then please promptly delete this e-mail and any attachment and all copies and inform the sender. Please note that any views or opinions presented in this email are solely those of the author and do not necessarily represent those of Robert Gordon University. Thank you.